As we continue to measure relative humidity in more and more environments with ever increasing accuracy demands, we are pushing the humble capacitive humidity sensor into new realms.
Accuracy, drift, operating range and chemical resistance are key challenges for the relative humidity sensor industry. Our sensor experts work hard to develop new polymers and construction methods to ensure the best performance. At the same time advanced electronics and probe housings enable digital calibration and complex temperature corrections to further increase accuracy and performance. A final and often neglected part of ensuring a relative humidity probes performance is its filter. The correct filter ensures fast response and environmental protection. Filters also offer mechanical protection and eliminate damage caused by extreme airflow.
However understanding why sensors fail is often difficult to predict or understand. In many cases the chemicals and contaminants that sensors are exposed to are unknown. In these situations often selecting the best sensor can only be achieved through mutual relationships built around quality support and service.
In the UK we have worked closely with many customers and in combination with our Swiss technical divisions to select and develop solutions for some highly aggressive and challenging environments. Some of these projects are examined below in more detail.
Hydrogen peroxide vapour sterilisation.
– Hydrogen peroxide vapour is used to chemically sterilise environments and products by generating a vapour of toxic Hydrogen Peroxide. When the humidity reaches the dew point of the surfaces condensation forms sterilising all surfaces. However the chemicals are also highly aggressive to humidity sensors and constant cycles of saturation worsen the effects.
– Making use of Rotronic’s specifically designed H2O2 resistant sensor as well as additional conformal coating to protect exposed connections in further combination with improved customer understanding around handling and storage, has resulted in a solution that has exceeded customer expectations. Importantly, whilst this was not achieved first time around, through a partnership driven towards the end goal we achieved success.Chemical degradation on the sensor surface Commercial composting.
– Accelerated commercial composting is an impressive sight to see. The chemical and biological processes occurring are complex and surprisingly aggressive. The wrong materials can literally become part of the final compost if you are not careful. Chemically resistant sensors help to provide some longevity to instruments but one of the key areas requiring extra attention is around cable and filter design. Modifying a standard industrial grade sensor with bio-resilient cables ensures the probes are not eaten alive!
Highly accelerated life testing.
– As a supplier to many chamber manufacturers and companies providing testing services this is a common application. Chambers are cycled between high and low temperatures and humidities to simulate many years aging over a short period of time. The same effects are happening to the humidity sensor – critical for the control or validation of the chamber conditions. Using industrial sensors with electronics isolated away from chamber space reduces the effects of the sudden changes. But also care taken placing the sensor away from humidity outlets and well into the chamber to avoid stem conduction all help to avoid the sensor becoming saturated as temperature cycle – which is one of the main causes for corrosion and drift. Finally, careful filter maintenance is always important.
Swimming pool monitoring and control.
– Our featured image shows chemical formation on a non-Rotronic sensors connections. Rotronic uses inert metals in the sensor design to reduce the re-activity of the sensor to chemicals in the environments. Swimming pools have a mix of high humidity, chemicals and high temperatures which work together to corrode unprotected electronics. Sensor location is key to avoid direct exposure to spray and neat chemicals. Suitable filters and if required chemical resistant sensors have proven highly successful where other instruments have failed.
So you can see not all applications are easy and we have not even begun to explore the basic issues of accurate measurement and control present with every humidity sensor installation. However our belief and aim is that through communication and partnerships we can provide the right product to ensure the desired mix of performance, resilience and price for our customers – it’s not easy but it makes life interesting!
Dr. Jeremy Wingate